Test method: Electrochemical Dräger sensor for 1/4“ technology; alcohol-specific.
Test range: 0 to 2.5 mg/L; display indication when exceeding test range.
Sampling: Standard: Automatic sampling when minimum volume is reached. Passive sampling or manual triggering of sampling possible.
Ready for use: Approx. 2 s after powering up.
Test result display: Approx. 3 s (with 0 mg/l), 10 s (at 0.5 mg/l, room temperature).
Operating temperature: -5 to +50 °C; -4 to +122 °F Display
LED: Graphic LCD display with background illumination; 41 mm x 24 mm (128 x 64 pixels) 3 colours to indicate messages or alerts
Acoustic alarms: Different signal tones indicate various messages and alerts
Memory: Space for saving the last 5000 tests with test number, date and time
Power supply: Two AA batteries or NiMH batteries, battery indicator on the display. Around 1500 breath tests can be conducted with one set of batteries. Charging rechargeable NiMH batteries is possible in device.
Calibration: Wet gas or dry gas calibration.
Dimensions (H x W x D), weight: 151 mm x 62 mm x 38 mm, approx. 195 g incl. batteries.
Device configuration: Device settings configurable directly via device menu (PIN required). No additional PC software required.
Vibration and impact: Meets EN 60068-2-6, EN 60068-2-27, EN 60068-2-64, MIL-STD 810F.
CE marking: Directive 89/336/EEC (electromagnetic compatibility).
Data interfaces: USB to PC; optical interface for printer control.
Standard: Meets EN 15964.
Protection type: IP 54.